Published: 2020-12-17 Article source:
GMT-P1, a basic weighing transmitter was launched last month, which can be used in variety industrial field. Its calibration without weights makes calibration process more convenient and efficient. Now, we are going to share with you how to calibrate GMT-P1.
There are two calibration method of GMT-P1:
Equipment used in Calibration
The calibration with weights can be divided in three steps:
For calibration without weights, there are 2 different modes can be chosen:
By historical data, you need to use the recorded millivolt data at zero point, the gained millivolt data and weight value. By theoretical data, you must ask the loadcell manufacturer to provide the accurate loadcell sensitivity value. By this data you can calibrate the weighing indicator in 1 minute but with high accuracy.