In general, analog output of weighing indicator will be calibrated and tested at the factory. When abnormal analog output is detected, we can calibrate to correct the analog output.
So, in this video, we illustrated how to calibrate and test the analog output of weighing indicator, taking GMT-P1 as an example.
GMT-P1, a basic weighing transmitter was launched last month, which can be used in variety industrial field. Its calibration without weights makes calibration process more convenient and efficient.
General Measure Intelligent Bag Filler is upgrading for better performance.
Free sample testing application of GMT-P1 has been opened globally.
Great sales achievement in Turkish market 2020 by Mr. Gurbuz Arici, with General Measure global expansion strategy - exclusive partnership.
To facilitate troubleshooting of indicator communication, GM has recorded a technical video as reference.